What is an Anti-Foulant Device? Does it affect the conductivity cell calibration? How often should I replace it? Does it require special handling?
The Anti-Foulant Device is an expendable device that is installed on each end of the conductivity cell, so that any water that enters the cell is treated. Anti-Foulant Devices are typically used with moored instruments (SBE 16, 16plus, 16plus-IM, 16plus V2, 16plus-IM V2, 37-SM, 37-SMP, 37-SMP-IDO, 37-SMP-ODO, 37-SI, 37-SIP, 37-SIP-IDO, 37-IM, 37-IMP, 37-IMP-IDO, 37-IMP-ODO), thermosalinographs (SBE 21 and 45), glider CTDs (Glider Payload CTD), moored profilers (SBE 52-MP), and drifters (SBE 41/41CP Argo float CTDs), and optionally with SBE 19plus, 19plus V2, and 49 profilers.
Anti-Foulant Devices have no effect on the calibration, because they do not affect the geometry of the conductivity cell in any way. The Anti-Foulant Devices are mounted at either end of the conductivity cell. For an in-depth explanation of how Sea-Bird makes the conductivity measurement, see Conductivity Sensors for Moored and Autonomous Operation.
Useful deployment life varies, depending on several factors. We recommend that customers consider more frequent anti-foulant replacement when high biological activity and strong current flow (greater dilution of the anti-foulant concentration) are present. Moored instruments in high growth and strong dilution environments have been known to obtain a few months of quality data, while drifters that operate in non-photic, less turbid deep ocean environments may achieve years of quality data. Experience may be the strongest determining factor in specific deployment environments. Sea-Bird recommends that you keep track of how long the devices have been deployed, to allow you to purchase and replace the devices when needed.
Note that the anti-foulant device does not actually dissolve, so there is no way to visually determine if the anti-foulant device is still effective.
The cost of the anti-foulant devices is small compared to the deployment costs, so we recommend that you replace the devices before each deployment. This will provide the maximum bio-fouling protection, resulting in long-term data quality.
Shelf Life and Storage: The best way to store Anti-Foulant Devices is in an air-tight, opaque container. The rate of release of anti-foulant is based on saturation of the environment. The anti-foulant will release until the environment is fully saturated (100% saturated) and then it will no longer release any anti-foulant. So if you keep Anti-Foulant Devices sealed well in an air-tight container, theoretically they will stay good for extended periods of time. Exposure to direct sunlight can also affect the release of anti-foulant; we recommend storage in an opaque container.
- For details, refer to the Material Safety Data Sheet, enclosed with the shipment and available on our MSDS page.
- Anti-Foulant Devices are not classified by the U.S. DOT or the IATA as hazardous material.